Rinsapen® I is an individual Pen-injector for insulin injections. Interval of 1 Unit (from 1 to 60 Units). Anti-glare dosage display. Easy-to-read dosage of insulin units chosen. Large print can be clearly seen from any angle. Dosage selector with “Shift Up Down” function with clearer click (best in the class) when setting a dose. Start push button to inject a dose with clear sound and tactile feedback. Light heat-resistant aluminum case provides for a smooth temperature mode of insulin in the cartridge. Mechanism of the cartridge holder prevents thread damage and provides for co-axial installation of the cartridge with a threaded rod – guarantee against piston jamming.
This drug is a prescribed medicine. It can be dispensed on a doctor's prescription only. There are contraindications. Please read the patient information leaflet. Do not use the medicinal agent if it was not prescribed by your treating physician. Always follow the patient information leaflet and your doctor’s recommendations.
Pen-injector is easy to use and recommended for independent use by adults and children over 12.
Rinsapen® I Pen-injector is intended for subcutaneous injection of insulin medicinal products and insulin analogues in 3-ml cartridges produced by GEROPHARM. The company is not responsible for damages occurring when using other cartridges.
Use Rinsapen® I Pen-injector with standard insulin needles intended for use with pen-injectors for administration of insulin and insulin analogues and registered at the territory of the Russian Federation. You can get further information on compatibility of needles from producers of the needles.
Cartridges and needles are not included into the delivery kit of Rinsapen® I.
Trade name: Rinsapen® I
Indications for use:
RinsaPen® I individual insulin pen-injector is intended for subcutaneous injection of insulin medicinal products and insulin analogues in 3-ml cartridges produced by GEROPHARM.
Attention! RinsaPen® Pen-injector is intended for individual use only!
Contradictions to independent use of the pen-injectors include:
Directions for use and dosage
to prevent injection of the wrong dose because of the air in the cartridge or a dirty needle. Replace the needle if necessary.
If you do not follow the patient information leaflet, there is a risk of wrong treatment, wrong dose, risk of infecting yourself or others.
If you have any doubts, see your doctor immediately.
Dosage form: put in the chart from the leaflet
Structure and elements of the pen-injector
Maintenance and storage
Ypsomed AG, Brunnmattstrasse 6, 3401 Burgdorf, Switzerland
The firm accepting claims
Lit. B, Degtyarny pereulok 11, Saint Petersburg, Russian Federation, 191144
Phone: (812) 703-79-75 (MLT), fax: (812) 703-79-76
Toll-free number: 8-800-333-43-76 (free call across the Russian Federation)
Marketing authorization: RU No. RZN 2016/5018 dated December 28, 2017
Creating a reusable pen GEROPHARM pursued the goal to take care of the patients – it was necessary to make a pen-injector maximally convenient for injection of the medicinal product.
Sound and tactile indication
Rinsapen® I has the following sound and tactile signals to control selection of an accurate dose:
when setting and correcting the dose, each rotating of the dose selector for 1 Unit comes with a click and effort.
The process of insulin administration comes with a sound that stops after administration of the full dose.
Limitations when selecting a dose
Maximum dose for a single injection is limited by the quantity of insulin in the cartridge. If there is not enough insulin in the cartridge for the dose selected, the dose selector will not rotate beyond the available quantity.
Do not use force to rotate the dose selector.
Identify and remember how much insulin is left in the cartridge and make an injection.
Replace the cartridge and the needle and check against the section “Check of efficiency and inflow of insulin” (page 7).
Set the dose selector at the value needed (lacking after previous injection) and administer the rest of insulin to get the full dose.
Installation of the cartridge partially used
Install the cartridge partially used into the cartridge holder.
Pressing with the finger, push the threaded rod fully to the stop. The threaded rod can also be pushed with the cartridge holder with a cartridge inserted into it.